Merits power chair troubleshooting
The scooter will not run. There are several reasons why a scooter may fail to operate - but the first thing to check is the battery. Check and make sure that the scooter's charging unit is connected to the scooter, then plug the charger cord into a standard outlet.
Continue charging the battery until the meter reads full. The scooter is fully charged but will not run. The key is not inserted correctly Insert the scooter key all the way into the keyhole and turn it to the ON position. Manual free-wheeling switch or lever Ensure that the manual freewheeling switch or lever is in the OFF position.
This switch or lever is present so that you can disengage the motor and braking systems if you ever need to push the scooter manually. Power-save mode has activated Remove the key from the scooter and wait for at least 30 seconds, then place the key back into the scooter. The power-save feature prevents the battery from being discharged while the scooter is not moving and while the key is in the vehicle. The activation time for this feature will differ based on your scooter type but generally ranges anywhere from 3 to 5 minutes.
Circuit breaker needs to be reset Push the main circuit breaker button to reset the motor. For clip-type batteries, ensure that both clips are connected fully to their respective battery terminals. Front-to-Rear Harness is not connected Ensure that the harness that connects the rear of the scooter is connected properly to the front section of the vehicle. This harness can become dislodged during normal use if you have not connected it properly during re-assembly.
The battery meter dips very low when I press the throttle. The scooter batteries might have a very low charge. You should charge your batteries fully before continuing to operate the scooter. You can do this by contacting the service center of the company where you purchased the scooter. The scooter hesitates or runs slowly even when I press the throttle fully.
There are a few reasons why this might be happening: Low battery charge The scooter batteries may have a very low charge. Operating on an upward slope Hesitation may occur if you are operating the scooter on an inclined surface.
Most scooters will run fine on a slight upward slope, but at a slower speed or not at all if the slope is too great. Scooter is at maximum capacity If the scooter is carrying a rider that close to its maximum capacity, the vehicle may run slightly slower than normal or not at all on an upward surface.
My circuit breaker repeatedly trips. This generally indicates a problem with your scooter batteries. You may need to charge them more frequently if they are getting older or have been in use for a long time. If this is the case, it would be a good idea to have them load tested for strength. Who do I call if none of these tips work?
Scootaround can be reached seven days a week at Outside of regular business hours, our technical staff is always on call for emergencies. Join our newsletter Sign up to receive information about Scootaround's latest promos, popular destinations, upcoming events, blog posts, and more!Troubleshooting the most common problems of lift chairs click here. If you need to locate a technician near you click here Verify that the battery pack is seated properly.
What can I do a mobility scooter or power wheelchair does power up but it does not move or it only moves a short distance? Make sure the free wheel lever is in drive mode if you can push the scooter or wheelchair and it easily moves then its in free wheel mode.
On the free wheel lever assembly you may have an electrical switch with wires going to the switch. Make sure those wires and switch are in proper condition. Verify that the circuit breaker has not been tripped and make sure the wires and the breaker are in good shape. Sometimes the breaker or connections are defective and will only work intermittently. Also, anything that that makes the motors work harder which will draw more current can trip the breaker like bad bearings, running on thick carpet, steep hills or the breaker can be defective itself and trips at lower amperage draws below spec than it would normally.
If it sounds funny or does not spring back correctly then check the assembly. When you check the assembly also make sure the wires and connectors are in proper condition. Pride Mobility Basic Troubleshooting Guides Shoprider Troubleshooting guide pdf, big file12meg. If you can, unplug as many cables as you can and visually check them to see if they look good if they look good then plug them back in.
Troubleshooting & Repair Instructions for Power Wheelchairs, Mobility Scooters, Lift Chairs ...
If the lift chair is still not working then you may have to find someone who knows how to use a multimeter in order to test voltages and or cable resistance connections. Handymen typically know how to use a multimeter and handymen will normally cost less in order to fix the chair. The battery voltage is greater than 32V. Check the battery condition and the connections. Suspect a charger malfunction. For a temporary solution, turn on the lighting system for a while in order to lower the battery voltage.
The controller has detected a shorted motor. Check the loom for shorts and check the motor. Contact your authorized dealer. Check the Free Wheeler Lever connections, the condition of the motor and loom s.
Return Thumb Control Levers to of neutral neutral and reset system. Contact your authorized dealer to readjust the wigwag to neutral if necessary.Quick Links.
Download this manual See also: Owner's Manual. Table of Contents. Wheelchair Merits p series Service Manual 15 pages. Service guide How to replace or repair the seat assembly To replace the seat body To replace the seat base plate assembly To replace the seat post Page 3: Index 1. If the powerchair crashed cause the armrest out of shape. How to replace or repair the body assembly The Body Assembly includes main shroud, fender, battery pack, footplate assembly, anti- tipper arm assembly - front, anti- tipper arm assembly - rear, caster wheel assembly, rear wheel assembly and the frame.
Page 7: To Replace The Fender 2. Please proceeding steps as follows if it has to replace. Replace the footplate assembly if it is worn out or deformed.
Hold the two screws 8. Page 9 loosen the nut 1. If the anti-tipper arm assembly - front is worn out or deformed. Page 11 washer if the anti-tipper arm assembly is worn out or deformed. Following the step above. The caster wheel assembly is composed of a PU wheel and fork. Replace the caster wheel assembly by the following steps if, after inspection, the assembly is found to be deformed. Page 14 Replace the driver wheel by the following steps if it is worn out or deformed.
EZ-GO / EZ-GO Deluxe
Follow preceding step above. If the PU tire is worn out. Hold the screw 5. This manual is also suitable for: P series P series. Print page 1 Print document 16 pages. Cancel Delete.Read Our Handy Guide.
We understand that your mobility equipment is an everyday necessity. We have factory trained service professionals that can provide service and repairs for your power chair or scooter at our authorized repair shop.
We also perform in-home service calls in the event that you are unable to transport your scooter or power chair to our store. We have put together an assistance guide which may save you time and money.How to Test Wheelchair Batteries Easy
Below is a picture with freewheel lever locations for some of out more common equipment. Do not use the mobility equipment while the drive motors are disengaged! Only engage the freewheel mode when on a level surface.
The lights will flash briefly and you will hear a beep sound. Press the horn button 2 times within 10 seconds to unlock the controller. First check your speed dial or power buttons to ensure they are at your desired speed settings. Are your batteries fully charged? Be sure the the connections to the batteries have not come lose. Try turning the mobility equipment off for an hour and then turning it back on.
It is a safety feature built in to the equipment which helps prevent it from burning out the motors due to overheating under strenuous usage. There is a normal clicking noise from the wheels when stopping and starting. This is the brakes engaging and disengaging when stopping and starting. The controls are sensitive and therefore require the joystick to be pushed straight for the unit to move forward. This is also true for reversing.
These issues may also be caused by one of the freewheel levers being disengaged. Re-engaging the freewheel levers may resolve the issue. Have you replaced the batteries yourself recently? Push the joystick forward and hold for about 2 seconds until you hear a chirp. Push the joystick backward and hold for about 2 seconds until you hear a chirp. Unlock Your Controller Turn mobility equipment ON Push the joystick forward and hold for about 2 seconds until you hear a chirp.
Release the joystick and you should hear a long beep tone, signaling that the equipment has been successfully unlocked.Display all pictures. Condition: New product. It works well indoors with it has a tight turning radius of 23". Read whole description. We use all major carriers to ship orders. Larger products are shipped by common carrier freight companies.
Unless White Glove Delivery is chosen, the client is responsible for removing the crating or box. Some minor assembly may be required. Add to cart. The minimum purchase order quantity for the product is 1. Titan Transportable Power Chair by Drive. PaceSaver Scout Boss 6. Click here for phone hours. Finance our products with Pay tomorrow. Click here to ask.
Reference: Condition: New product. Limited special offer. No, Thanks Yes, Please. Footplate Standard Elevating Leg Rests.
This product is no longer in stock. Availability date:.
Select options for product. Oxygen Holder. Seat Cushion. Back Cushion. Front Riggings. Weight Capacity lbs. No customer questions for the moment. Ask a question. Need help? Call us 1- Click here for phone hours.
The PD by Merits Health has many features that support what our customers are saying. First the seat can elevate a total of 7. This is important for those who struggle to reach higher places within the home such as counter tops.
What is really nice is the ability to ride around at half speed safely within your home or on the go at 2. At 5 MPH which is the maximum speed, the chair must be at the lowest position to accomplish the maximum speed. The Merits PD is a very comfortable wheelchair to use for everyday outings. The seat which is 20" wide included a comfortable and ergonomic captains chair with headrest. The seat is adjustable in depth from 18" to The Merits PD power wheelchair with elevating seat offers a full suspension system for optimum stability, durability, and reliability.
The Merits PD tires and casters are all solid and durable rubber insuring you, you will not get a flat tire while on the go. The Merits PD is equipped with stellar inline motors that measure 24v and watts. The electromagnetic and regenerative brakes allow your machine to stop when you want it to stop.
This controller precisely controls two inline motors that are rated at 12v and watts. This quality power wheelchair with elevating seat has a weight capacity of pounds. It also features a footplate that will hold up to pounds without the chair tipping forward. This is convenient to those who have a hard time flipping the footplate up. Here at Mobility Scooters Direct our customers get the benefit from buying from us than any other competitor.
With our customer service and excellent product knowledge we will be there for you for your questions and concerns. We are a real people company not ran just by computers.
Buying the Merits PD power wheelchair from Mobility Scooters Direct will provide you a free array of accessories including a cup holder, armrest bag, weather cover and rear basket. This is also important than buying from retail, which our cost is less than half the cost of retail. Finally another strong point is knowing that your product has not sat in a showroom for a long period of time, which may void the warranty by the time you purchase it.
When buying from Mobility Scooters Direct, your product comes to you untouched by anyone other than a Merits Technician who has thoroughly inspected and tested each unit according to FDA guidelines. If you are looking for a power wheelchair liftthen make sure to check those out for an opportunity to take you chair on the go. Corporation warrants to the original purchaser of this wheelchair product that it is free of defect in material and workmanship and that, when operated within the guidelines and restrictions of this manual, will remain so free of defect in material and workmanship for a period of One 1 year from the original date of purchase.There is certain criteria that must be met for Medicare to pay for any medical equipment.
It must be considered medically necessary for the patient and all of the following criteria must be met:. Your condition is such that a wheelchair is required for you to get around in your home and you are unable to operate a manual wheelchair. You are capable of safely operating the controls of the scooter or power chair. And, in addition, a patient who requires a scooter usually cannot perform functional ambulation skills and has limited strength or use of the upper extremities.
Note: An individual who requires a power wheelchair is totally non-ambulatory and has severe weakness of the upper extremities that prevents them from self-propelling in a manual wheelchair. Pilot Stair Lift. Avid Rehab. Power Wheelchairs. Mobility Scooters Scooter Accessories. Patient Aids. Beds and Related Products. Manual Wheelchairs. Wheelchairs Accessories. To order parts for your Merits product, please contact the dealer the product was originally purchased from. If you are unable to locate the dealer, please call for further assistance locating a dealer in your area.
My bed is not working properly. The head or foot sections will not go up and down. Make sure the power cord is plugged into wall outlet and junction box at foot of bed. The standard fill pressure of a full cylinder is psi.